Four screw holes and linear guidelines in the front plate enable usage of customizable fixtures to accurately position your samples – and always in a repeatable manner.
Camera previewer for accurate sample positioning
The integrated camera viewer provides a clear preview of the sample for perfect targeting and control the area to be measured. Capture images of the area measured for more accurate reporting.
Reflectance measurements, horizontal or top-port
The CM-36d can measure solid, powdery or liquid samples in reflectance or transmission mode, horizontally or upside down, thus offering great flexibility to measure all kind of samples.
Easy to read tilted LED indicator panel with measurement button
The tilted LED indicator panel shows you all your settings at a glance and what status your device is in. In addition, the measurement button in the panel can be used to measure samples remotely while holding them to the front plate.
Wavelength Analysis & Adjustment – “WAA”
“Analyze - Adjust - Activate” - Konica Minolta’s patented technology to check and adjust wavelength shift during each instrument calibration procedure assures unsurpassed long-term measurement stability and keep measurements stable even for changes in temperature or environment over time. WAA lets you easily distinguish whether the instrument itself is the cause of a problem or not, without starting guesswork or faulty assumptions
각 종 분야의 연구, 개발 부문, 품질관리 부문
Benchtop Spectrophotometers CM-36d
Illumination/ viewing system
di:8°, de:8° (diffused illumination, 8° viewing)
Conforms to CIE No.15 (2004), ISO7724/1, ASTM E1164, DIN 5033 Teil7, JIS Z 8722 Condition c standard.
Integrating sphere size
Dual 40-element silicon photodiode array
Spectral separation device
Planar diffraction grating
approx. 10 nm
1 Pulsed Xenon lamp (UV full only)
Light source life time
more than 1 Million
Illumination / Measurement area [mm]
LAV： Ø25.4 / Ø30.0
MAV： Ø 8.0 / Ø11.0
SAV： Ø 4.0 / Ø 7.0
approx. 0.7sec (SCI or SCE)
Min. measurement interval
approx. 1.5sec (SCI or SCE)
Std. dev. within ΔE*ab 0.02
within ΔE*ab 0.15 (LAV/SCI) Average for 12 BCRA Series II color tiles compared to values measured with master body.